Important notes for installation
- Cables must be fitted through 1/2″ NPT cable gland or ATEX conduit (not supplied with instrument).
- Make sure after cabling to close tight cover and cable gland, in order to keep IP66 rating (IP65 with option OPV, overpressure relief valve).
- Open cover only after de-energizing instrument.
- Attention: Check local safety rules and warnings on unit and manual for a correct use of the instrument in hazardous area.